I am not saying that you should not purchase a Hickok like a or at the right price and condition, however in the absence of other considerations, these are good choices. Model A. This is the best of the basic early testers. Data is available. Has sockets up to 9-pin mini.
|Country:||Bosnia & Herzegovina|
|Published (Last):||15 July 2007|
|PDF File Size:||19.48 Mb|
|ePub File Size:||5.17 Mb|
|Price:||Free* [*Free Regsitration Required]|
One of my favorite artifacts at the Pavek is this Hickok classroom demonstrator, donated in by Dr. Frederick Koelsch, a retired chemistry professor at the U of M.
The top meter on Dr. The meter on the left shows the plate current when grid voltage is most negative, while the meter at the right shows plate current with the grid signal voltage most positive. The meter at the bottom of the panel shows the difference between the two states and is calibrated in micromhos. Hickok demonstrator Much has been written about the circuit, designed by Job R. Barnhart in It allowed Hickok to produce the best-selling dynamic mutual conductance tube tester on the market until the patent expired in Dynamic Mutual Conductance Gm is the measure of change in plate current produced by a change in grid voltage.
It then puts a known AC signal on the control grid and measures the difference between the plate currents produced by the changing grid voltages. The true dynamic mutual conductance is then displayed on the direct reading meter. The keys to the performance of the Hickok circuit are accurately calibrated potentiometers, a sensitive and accurate Gm meter, and the absolute predictability of the phase relationships of the voltages produced by the secondary windings of the power transformer.
The image on the left illustrates the Hickok design in its simplest form. The cathode of the tube under test is tied to the negative lead of the volt plate supply and to the wiper of the bias control. The control grid is tied the the negative side of the volt screen supply.
The wiper of the bias pot is shown closest to the negative end of the voltage divider across the screen supply. At this point the cathode and the grid are at the same potential, in other words, zero grid bias. Here the wiper of the bias control is shown closest to the positive end of the voltage divider. As the wiper on the bias pot is adjusted toward R the cathode becomes positive in relation to the grid. Grid voltage is always measured in relation to the cathode.
If a five-volt transformer winding is added between the cathode and the bias pot, then the grid will swing five volts above and five volts below whatever voltage is dialed in on the bias pot. When P2 of the 83 is positive at the same time that the grid end of Secondary 5 is high, the Gm meter will want to swing toward the high end of the scale. When P1 of the 83 is positive at the same time that the cathode end of Secondary 5 is high, the Gm meter will want to swing toward the low end of the scale.